ANESTHETIC GAS SCAVENGING DISPOSAL SYSTEM PUMPS (AGSS/WAGS)
This integrated system is used to evacuate the anaesthetic gas breathed out by the patients in the operating rooms or post-operating awakening rooms. The system is manufactured in compliance with ISO 7396-2 regulatory standard.
The vacuum level is adjusted by the proper regulation valve and maintained stabilized by the intermittent works of the pumps. The transducer read and shows continuously the generated vacuum level.
REMOTE CONTROL MODULE
This module, properly connected to the AGSS delta P system and installed in proximity of the operating rooms, allows to start/stop the AGSS system and control that everything is working correctly by the specific status and alarm LEDs. It is possible to connect a supplementary vacuum switch to monitor locally the system, for example after switch-off valves or vacuum regulators.
It is possible to install in parallel scheme more modules, the AGSS system will work if at least one of the switch is in ON position. The silencing pushbutton allows to cut off the ringer in case of alarm, if after 15 minutes the alarm conditions are still present the ringer restart automatically.
- Electric board (local control panel);
- Two vacuum pumps with lateral channels with double parallel flow
- One pressure tranducer
- Vacuum regulation valve;
- Remote control system module;
- Two Aerating filters;
- Maximum silence
- Minimum overall dimensions
- Safety and reliability
- Fully complying to ISO 7396-2 regulatory standard